The Calorimeter Temperature Sensor
The event of your C-3500 flame depth calorimeter was primarily for the measurement of combustion chamber heat fluxes in electricity creating facilities. The calorimeter temperature sensor enables the immediate and instantaneous electric power analysis of combustion chamber gasses for very best stoichiometric fuel/air combine. On top of that, the calorimeter is greatly Utilized in verifying warmth source intensities at various cracking tower stations along the complete oil refinery process, which assures procedure repeatability. Another crucial software is applied to the flame screening of resources and components under Federal Aviation Laws. Other applications incorporate electrical power generation, hotbox flame control, jet exhaust output, all-natural gasoline Vitality material, reactor output, furnace depth and several ASTM flame test evaluations.
The flame intensity temperature sensor is actually a one.00 – 2.00 inch diameter, rod that is definitely positioned in immediate contact with the heated gasses. The signal created is directly proportional on the thermowell nearby warmth flux; a radial traverse of the supply yields a measure of the total electricity output. The sensor yields D.C. millivolt indicators that may be calculated with regular millivolt meters or recorders.
How the Calorimeter Performs
The C-3500-3600 Flame Calorimeter should be to be inserted via a port instantly in the combustion fuel stream. The HT-50 High Temperature Warmth Flux Transducer has actually been welded and warmth sunk into a area 50mm through the idea of the C-3500 Probe(C-3600 has sensor at tip of probe). The placement of the sensor is on precisely the same side because the h2o discharge port and continues to be quickly marked for your usefulness. This place need to be famous and located so the combustion gasses impinge immediately upon it. The greatest detected fluxes will occur when the sensor locale is at 900 or in a stagnation issue into the flame front.The calorimeter includes an internal drinking water feeding tube in the hollow cylindrical probe. Awesome drinking water enters with the interior tube and flows out in the annular gap concerning the inner tube and interior wall with the outer probe tube. A Management method varies the rate of movement of h2o to take care of the inside temperature in the probe wall at a constant price. Water coolant need to be provided into the probes rear inlet at a movement price of ten – twenty Liters/Minute. Boiling from the cooling h2o will have to never be allowed, for very substantial thermal flux fees, the water/coolant flow amount have to be greater till the exit water temp is tepid. The typical warmth flux is calculated by multiplying the C-3500-3600 Calibration Consistent from the measured DC micro-voltage.